Firstly a merry christmas to all.
My resmed airsense 10 autoset stopped working in the early hours of this morning.
My machine was bought just on a year ago in the US.
I suspect that the ac adaptor is bust although the power plug is on...no light comes on when the unit is on
Can someone advise me if this i a common occurence given that is a year old.
I would appreciate some advice before consulting the place i bought it from
If you have a voltmeter, or know someone who has one, it would be fairly easy to measure the voltage coming out of the plug that goes into the A10. Just plug the ac adapter into the wall, but not into the PAP machine, and then use the voltmeter to measure across the plug. If there is no voltage, or it is more than a few volts lower than the 24V it is supposed to be, the adapter is kaput.
It's not a common thing but it is an electrical device. Subject to failure.
Another ResMed adapter will set you back about ~$65
Using FlashAir W-03 SD card in machine. Access through wifi with FlashPAP or Sleep Master utilities.
I wanted to learn Binary so I enrolled in Binary 101. I seemed to have missed the first four courses.