Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
[CPAP] error code e-23
04-06-2012, 10:11 PM
HELP!!! I have a REMstar Pro with C-Flex and a RP-REM Heated Humid Dom Humidifier. I am getting this code e-23 and don't know how or if it can be reset. Should I open it and fix it or is it the electronic board ?? I have no insurance and have been out of work for 4 years and am going to school so if I have to get another, I'm screwed.
04-06-2012, 10:25 PM
According to an older post, code E-23 is:
Quote:E-23 ERR_INVALID_DRIFT_INDEX_AT_CAL •Improper calibration •Occluded or restricted pressure tube •Defective flow sensor •Recalibrate •Replace pressure tube •Replace Main PCA
04-07-2012, 12:02 AM
(04-06-2012, 10:11 PM)rollsroyce Wrote: HELP!!! I have a REMstar Pro with C-Flex and a RP-REM Heated Humid Dom Humidifier. I am getting this code e-23 and don't know how or if it can be reset. Should I open it and fix it or is it the electronic board ?? I have no insurance and have been out of work for 4 years and am going to school so if I have to get another, I'm screwed.
You may find help in this thread
Breathing keeps you alive. And PAP helps keep you breathing!
04-07-2012, 07:08 AM
Yep, take a look that the thread that Jumpstart listed. Most likely, on these older REMstar machines what's happening is that the small rubber tubing inside the machine (see the pictures in that thread) are becoming old and brittle, and they will develop a leak.
Check out this from a previous thread:
Quote:Sleepster offered a good solution and sometimes completely un-plugging and re-starting the machine will help... however if it doesn't help, look at zonk's solution.
Apnea Board Administrator
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
|Possibly Related Threads...|
|need CPP or CPT code for resmed aircurve 10 ASV with oxymiter||xxyzx||16||182||
9 hours ago
Last Post: holden4th
|[CPAP] Res Med S9 help! Error: Settings reset to default values, please contact service pr||dexter020281||16||11,545||
04-15-2017, 10:05 PM
Last Post: SuperSleeper
|[Equipment] ResMed Autoset II w/EPR and humidifier -ERROR||thisazlife||2||177||
04-12-2017, 03:29 PM
Last Post: thisazlife
|Philips Respironcs Remstar BiPAP error code E48||dcubed||6||197||
04-03-2017, 10:07 PM
Last Post: Sleeprider
|[CPAP] resmed s9 auto set error message high temperature fault refer to owners manual||beebsdaddy49||2||180||
03-10-2017, 07:49 PM
Last Post: trish6hundred
|ResMed S9 Humidifer 5i Error Message||atta_gurrrrl||4||3,155||
02-05-2017, 07:56 AM
Last Post: gjw
|SD Card Error (Nothing new)||justMongo||3||290||
02-02-2017, 10:22 AM
Last Post: Crimson Nape