Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
Default Settings? - Respironics PR System One 60 Series REMstar Pro with C-flex+
08-24-2013, 05:31 AM
What are the default settings in "Setup" for a Respironics PR System One 60 Series REMstar Pro with C-flex+ with a smaller diameter heated tubing?
(08-24-2013, 05:31 AM)edspyhill09 Wrote: What are the default settings in "Setup" for a Respironics PR System One 60 Series REMstar Pro with C-flex+ with a smaller diameter heated tubing?From the manual
• Tubing type - This setting allows you to select the correct size diameter tubing that you are using with the device. You can choose either (22) for the Philips Respironics 22 mm tubing, or (15) for the optional Philips Respironics 15 mm tubing.
When using Heated Tubing, the device will automatically change this setting to the appropriate tubing type (15H)
The manual is available via email (scroll down to section three to ask for the manual)
11-29-2013, 02:57 PM
(08-24-2013, 03:09 PM)zonk Wrote:(08-24-2013, 05:31 AM)edspyhill09 Wrote: What are the default settings in "Setup" for a Respironics PR System One 60 Series REMstar Pro with C-flex+ with a smaller diameter heated tubing?From the manual
Update, many months later. I finally was able to resolve several problems. The first being the CPAP company was somewhat incompetent.
As to the machine: I was sent a heated tube which is different than a standard tube. I still could not get the air to be heated properly. Then I found out the power supply was not powerful enough to heat the tube. So I was given an 80 watt power supply. That fixed the problem .
I had lowered the pressure to 8 when the air flow was too cold. Then I slowly increased the air pressure from 8 to the current 11. I will go to 11.5 then finally to the 12 that is prescribed.
I am sleeping with the CPAP mask on for 6 to 7 hours a night. I'm actually at the point where I prefer sleeping with the CPAP mask on. I even take on short vacations.
|Possibly Related Threads...|
|Respironics Bipap Auto SV settings ??||ecojedi||5||69||
1 hour ago
Last Post: robertbuckley
|[Equipment] Phillips Respironics BiPAP autoSV||lookingforbettersleep||7||172||
02-19-2017, 10:26 PM
Last Post: Sleeprider
|Can't seem to dial my settings in||C0mbe||2||119||
02-18-2017, 01:32 PM
Last Post: C0mbe
|[CPAP] Res Med S9 help! Error: Settings reset to default values, please contact service pr||dexter020281||14||10,816||
02-18-2017, 01:22 AM
Last Post: vsheline
|How to access Airsense 10 Expiratory pressure relief (EPR™) settings||CCDrogan||3||141||
02-17-2017, 11:30 PM
Last Post: trish6hundred
|Help with respironics system one||Philipg33||11||273||
02-17-2017, 12:22 AM
Last Post: Ghost1958
|[Equipment] How to properly set VAuto clinician settings?||Jioaa1988||4||176||
02-08-2017, 09:35 AM
Last Post: Sleeprider