02-06-2016, 06:38 PM
(This post was last modified: 02-06-2016, 07:27 PM by vsheline.)
(02-05-2016, 03:58 PM)Sleepster Wrote:
Quote:IMPORTANT FUNCTION NOTE: The AirCurve 10 VAuto machine does spontaneously regulate the breathing pattern to avoid cycling which results in inefficient breathing. This feature CANNOT be turned off. This machine is NOT a BiLevel ST machine and DOES NOT have a back up rate setting.
By the way, when we were looking at these machines online she went to Supplier #1 web site and I had her read the above quote from the A10 product description. She said it doesn't make sense to her, either.
The word "cycling" here refers to the end of inhalation / beginning of exhalation.
In other words, I think ResMed is saying the machine avoids prematurely ending Pressure Support.
I think the implication is that cycling back to the low EPAP pressure is delayed slightly if our inhalation is very short or stop-and-go. Or can be delayed, by appropriate adjustment of the minimum time allowed for inhalation.
Not sure why there is such strong emphasis on the inability to turn this feature off, since I suspect the S9 VPAP Auto and AirCurve 10 Auto have the same adjustability in this regard.
Membership in the Advisory Member group should not be understood as in any way implying medical expertise or qualification for advising Sleep Apnea patients concerning their treatment. The Advisory Member group provides advice and suggestions to Apnea Board administrators and staff on matters concerning Apnea Board operation and administrative policies - not on matters concerning treatment for Sleep Apnea. I think it is now too late to change the name of the group but I think Voting Member group would perhaps have been a more descriptive name for the group.
I believe what they are saying is that the machine allows a programmable window for you to spontaneously cycle to EPAP. It does not provide a timed backup rate Like the VPAP ST. This would be the same as our S9 VPAP Auto. At least that is my interpretation of their clumsily worded statement.