Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
[Equipment] Blower will not come on
01-19-2016, 12:46 PM
Hi am an new here and have a delima. I am a medical missionary working and living in the remote villages of Guatemala. I have a REMstar plus model M CPAP. I used it for several years and loved it. I had to move to an area with no electricity so could not use a CPAP. Now I have electricity and went to use my CPAP and everything seems to come on...( screen has no service messages or errors) But blower just never comes on. Do you all think it is just gone, or does it need some kind of reset. I do not have an option of getting a new one or taking it to a tech as we have none here in Guatemala. Any help appreciated.
01-19-2016, 02:02 PM
It's possible that during the time it was sitting that the blower bearings have "stuck."
There is no way to repair it that I know of.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
If you have access to compressed air, you could try directing a short blast into the hose so it will pass backward through the unit. Remove the filter first, in order to maximize the possibility of success. That could free the blower wheel bearing if it's merely hung up due to congealed lubricant. If it gets started, it might do well for quite a while.
If no compressed air, try blowing a few big puffs by mouth. Worth a try.
01-19-2016, 04:07 PM
(01-19-2016, 12:46 PM)Junglemedic Wrote: I do not have an option of getting a new one or taking it to a tech as we have none here in Guatemala. Any help appreciated.You can get a reasonably priced new or used machine from Supplier #2, ship internationally
Machine choices http://www.apneaboard.com/wiki/index.php...ne_Choices
01-19-2016, 07:31 PM
WELCOME! to the forum.!
Hang in there for more responses to your post and I hope you can get your machine working.
|Possibly Related Threads...|
|[Equipment] Respironics blower failure||SleepyinMTL||2||390||
12-03-2016, 12:52 AM
Last Post: trish6hundred
|[Equipment] Blower Hours||duffer54||6||531||
10-10-2016, 11:21 PM
Last Post: trish6hundred
|Respironics System One A-Flex, blower won’t start, Beep Beep Alarm, “Service Required||rae95131||4||1,491||
07-05-2016, 11:42 PM
Last Post: Ghost1958
|Backup cpap, buy blower or blower + humidifier?||IDRIck||5||600||
06-15-2016, 08:47 PM
Last Post: Mosquitobait
|Not a Noob but new blower now large leaks||zload||8||1,993||
05-28-2015, 04:09 PM
Last Post: PaytonA
|leaf blower on my face||CArmistead||10||2,451||
04-28-2015, 05:35 PM
Last Post: truetopath
|Replacement blower motor||OddBHavier||7||5,295||
04-21-2015, 12:47 PM
Last Post: Terry