Has been a week or two since I last lurked but i want to share my sucess with my Airsense 10 issues and how I hope they are now solved.
So in summary my issues were;
Blocked tube alarms
Periods of no data being recorded
Fast pressure ramps making exhale tricky.
Well i finally managed to get some feedback from Resmed directly here in Australia and while some mysteries remain, so far so good!
The issues are all related and are all to do with the P10 pillows and excess humidity.
So as you know the P10 has very fine mesh vents. These are easy to block and as reported elseweher on here, even the surface tension of water can cause restriction. So if the meshes get congested , the machine reports tube blocked and to tries and cures this by immediately ramping pressure to try and clear blockage. As its now in this recovery mode, it stops recording data, even though it keeps working!
So what causes this?
Well could be blocked mesh due to dust or lack of cleaning. I reckon I was good at that, Id water rinse every day and gentle brush once a week. I wouldn't use soap but now do - a dermalogical liquid one and I'm very careful to rinse thoroughly.
Next was was humidity (and this is where it gets a little odd). I always had climate control in auto and this problem developed. Some times I noted moisture in the mesh but never noted as a problem. It didnt have chronic water dropong out in the tube. Accordindig to Resmed, unsurprisingly in auto the unit should set humidity to 85% and so should never see condensation. I should also add Im in Australia so room temps even at night are warm.
Anyway decided to switch to manual control. Tube at 27C and started with humidity at 4 and had proper condensation, dropped to 3 and all seemed good but still got error events, now down at 2 and so far so good - no problems. Now Resmed felt 2 might be too dry but its ok for me.
So if you have similar issues to me, try swicthing to manual and dropping humidity. They also suggested upping temp to 28 or 29 C but I haven't done this.
Good luck buys and I hope this helps soemone. im so relieved getting better sleeps and the important data for compliance.