Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
[Equipment] Need Help w/ ResMed M Series
11-12-2013, 03:10 PM
Whole unit leaks water, but it HASN'T been tilted or bumped! It doesn't appear to be coming from the pull-out reservoir. 2 nights ago the blue LED on the humifier's humidity setting knob started blinking, and won't stop unless the unit is unplugged. And, I believe the humidifier's heater stopped working. I assume the heater and blinking issues are due to the water leak, but without a manual, it's hard to be sure. And unless there's a local problem with gravity, or somebody started moving it in their sleep, I can't figure out how the water is leaking (it sits on a very stable nightstand, and I don't fill the reservoir nearly full). Thoughts?
11-12-2013, 03:23 PM
Hi wtdavis55, welcome to the forum
There is no ResMed M Series, the M series is Respironcis machine
Look at the identification pictures and tell us which machine do you have so people who use this type of machine can helps
11-12-2013, 03:31 PM
WELCOME! to the forum.!
Hang in there for more suggestions.
11-17-2013, 07:31 PM
Maybe the tank is leaking where the metal plate seals to the plastic. It might only leak when it's in the machine being pressed on by the mechanism, and being heated.
The first step would be to replace the water tank.
However, the blinking blue light is usually a fatal problem with the heater in the humidifier.
You might try draining the water tank and running it with no water for a few nights to be sure that the humidifier is thoroughly dried out.
Don't spend too much money fixing the old machine. Check my signature line for a much better new model machine and buy it online or from someone on craigslist.
|Possibly Related Threads...|
|Would 2 12V batteries in series avoid need for Resmed''s (A10) converter||2SleepBetta||1||123||
03-16-2017, 09:39 PM
Last Post: Sleeprider
|[CPAP] Respironics REMStar System One Series 60 Repair||lempface||6||456||
02-04-2017, 07:01 AM
Last Post: cate1898
|Remstar Auto M Series "Service Required"||msgill59||4||550||
01-22-2017, 08:57 AM
Last Post: msgill59
|[Equipment] Annoying on/off beep on Respironics BIPAP S/T C series||rcattell||8||659||
11-06-2016, 12:07 PM
Last Post: rcattell
|[Equipment] REMstarPro M series||Lorelei||9||512||
08-19-2016, 11:36 AM
Last Post: OpalRose
|Philips series 50 Pro CFlex + and series 60 series differences||stephennic||1||262||
08-16-2016, 09:34 AM
Last Post: Sleeprider
|[News] Philips-Respironics "DreamStation" - new CPAP series on the way||SuperSleeper||126||54,495||
08-10-2016, 05:31 PM
Last Post: Perchas