Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
[Equipment] Resmed S8 Compact Disassembly / Cleaning
08-15-2014, 04:26 PM
My S8 Compact is in need of having its impeller cleaned. When it starts up, it sounds like a siren. It got so loud it woke me up last night. Blowing compressed air through it helps quiet it down, but not enough, and over time it gets louder again. Changing the hypoallergenic filter doesn't help. I'd like to clean the impeller. But after removing the 4 Torx screws from the underside, I still can't get it open to access the blower impeller. Can anyone tell me how to proceed to open it up?
08-15-2014, 04:32 PM
Somewhere there are disassembed photos of an S8 into its major sub-assemblies.
Likely S8 repair is undertaken by replacing sub-assemblies.
I don't think you can get to the impeller.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
08-15-2014, 05:06 PM
WELCOME! to the forum.!
Hang in there for more suggestions and best of luck to you.
08-15-2014, 07:54 PM
(08-15-2014, 04:32 PM)justMongo Wrote: Somewhere there are disassembed photos of an S8 into its major sub-assemblies.
I'd like to try.....They put it together, right? So somehow (hopefully not destructively) it can be taken apart. I just need to know what hidden fastener scheme they used, so I can get it apart and still put it back together when I'm done. I've seen many hidden screws under labels, rubber feet, etc. But the bottom layer here seems very thick, and hard to remove if that's the key.
Thanks to the way JustMongo worded his post, my subsequent Google search for "disassembled photos of an S8" led me to THIS post right back here on this forum, in which there is a current link to the registered member files.
In there I found the Service Manual for an S8 Auto 25, which is close enough to my S8 Compact to reveal the secret to splitting the case after the 4 torx screws are removed from the bottom. It says " a great deal of force may be required to separate the cases". OK, 800 pound gorilla approved for action.
The deal is that the innards are all rubber lined. The rubber coating seems to want to adhere strongly to the plastic case until you force it apart for the first time. Once it finally starts coming loose it was easy to finish removing it.
Once you get inside, the S8 Compact is made of a lot of easy to disassemble and clean parts! Even the impeller cage is a 2 piece affair that opens up. So I was able to give everything a thorough cleaning to remove the dust that had gotten in over the past several years. It is much quieter running now, but still a bit noisy compared to when it was new. Not sure what else could be done. It still makes a faint siren sort of whine when running. I'm due for a new CPAP machine this fall anyway through my insurance, so it should get me by until then.
I'd never have taken the leap of faith to get this open if not for the information in this forum, as I was afraid there was something else inside holding it together, and I did not want to break it. So thanks to you all for helping me solve the problem!
(08-15-2014, 05:06 PM)trish6hundred Wrote: Hi Dreamweavil,
Thank you Trish! In fact I did find the solution and have solved my problem. Have a great night!
|Possibly Related Threads...|
01-24-2017, 06:41 PM
Last Post: joelrk35
|[Equipment] Cleaning apnea equipment||lapopessa||8||632||
01-23-2017, 07:20 PM
Last Post: Goldie54914
|Cleaning the Dreamstation reusable dark blue filter||KCap||5||484||
01-10-2017, 04:20 AM
Last Post: Ockrocket
|Cleaning Dream wear mask||Judy1843||3||451||
12-02-2016, 07:13 PM
Last Post: cate1898
|Cleaning Climate-control tube on ResMed Air Sense 10||GL Bishop||18||13,504||
11-07-2016, 03:31 PM
Last Post: Jonesie
|Water in hose after cleaning...||JustMac||3||561||
10-30-2016, 06:51 AM
Last Post: OpalRose
|CPAP cleaning system||Tommy C||97||69,208||
10-20-2016, 05:35 PM
Last Post: PaytonA