Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
[Equipment] Resmed S9 fan noise when inhaling
07-12-2016, 08:01 AM
My equipment is about 4 years old and recently started making a weird whine when I inhale. It seems like the fan speeds up on inhaling and normalizes on exhale. The pitch is just high enough to be annoying and is affecting my sleep. Is this normal wear and tear or do I need to have the unit serviced?
07-12-2016, 11:24 AM
It is normal for these machines to speed up when you inhale: they need to push through more air to maintain the set pressure. When you exhale, less is needed.
The new whine could indicate a few things. If it sounds more like a whistle, it could be a gasket that is worn out, not seated correctly, or pinched. Or that could be a hole in you hose. If the whine sounds more metallic, It could be a bearing in the blower (or whatever they use).
07-14-2016, 05:50 AM
WELCOME! to the forum.!
Hang in there for more answers to your question and good luck to you as you continue your CPAP therapy.
07-14-2016, 01:32 PM
I've also noticed my machine is noisier when I use my FFM and has gotten noisier as it gets older. I can switch to the nasal pillows and it not make much noise at all.
Disconnect the humidifier and connect straight to the blower unit. If the noise stops, then the racket is probably the humidifier gasket or water in the tub (I sometimes get a whistle if there is water on top the tub).
If the noise continues, then it could be bearings, an internal gasket, or just plain noise. Try putting a thick towel over the machine or put it in a drawer. Just ensure the filter is clear!! I keep a towel to toss over mine for those nights when, for some reason, every little sound irritates me and I can't sleep.
Apnea Board Moderator
Breathe deeply and count to zen.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
|Possibly Related Threads...|
|[CPAP] Whining siren noise and bearing replacement/repair||clintga||24||7,337||
02-16-2017, 06:24 PM
Last Post: Eppy
|[Equipment] ResMed Airsense 10 noticeable noise||hungrycactus||5||360||
02-10-2017, 05:00 AM
Last Post: nanbar
|[Diagnosis] Popping Noise||clark77494||5||389||
01-19-2017, 08:55 PM
Last Post: trish6hundred
|[CPAP] ResMed AirSense 10 makes loud noise when inhaling.||sanger||91||35,377||
12-31-2016, 02:00 PM
Last Post: PaytonA
|[Equipment] Machine makes groaning noise on inhale||wideawake34||5||559||
12-25-2016, 06:50 PM
Last Post: icyura10
|swift mask 'leak' noise--but not leaking...||hegel||15||1,159||
12-14-2016, 04:12 PM
Last Post: Sleeprider
|Clicking noise from CPAP when breathing||Fedman||6||507||
12-11-2016, 04:36 PM
Last Post: PaytonA