Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
[Equipment] system error
08-17-2014, 04:33 PM
Can I fix the "system error" on my RESMED cpap machine. It will not turn on.
08-17-2014, 04:48 PM
Pull the SD card out and leave it out when booting the machine up. If things go well and the machine starts up without a peep, get a new SD card and try it. If not, you may have to take it in for service.
If taking it in isn't handy or if you're paying for it out of pocket, there are some things to try before capitulating. You can try unplugging the machine and waiting 30 seconds +- and powering it up again. While it's powered down, hold all the buttons down just in case that may help.
If that doesn't work, you can reset the machine to default condition by entering clinical mode and opting for that from the menu offered.
If none of that worked, you're out of luck pal and will need to take to a technician for troubleshooting and repair or else use it for a door stop.
08-17-2014, 04:48 PM
WELCOME! to the forum.!
Which Resmed machine do you have? The name is next to the power button.
Hang in there and someone will be along to help you, soon.
Best of luck to you.
(08-17-2014, 04:33 PM)tfitzge1 Wrote: Can I fix the "system error" on my RESMED cpap machine. It will not turn on.If the machine is an S8, according to the manual ... component failure ..... record error number and contact the ResMed service center
Check Supplier #2 for used and new machines
Suppliers List http://www.apneaboard.com/forums/Thread-...plier-List
Archangle:Machine Choices http://www.apneaboard.com/wiki/index.php...ne_Choices
|Possibly Related Threads...|
|[CPAP] Res Med S9 help! Error: Settings reset to default values, please contact service pr||dexter020281||14||10,817||
02-18-2017, 01:22 AM
Last Post: vsheline
|Help with respironics system one||Philipg33||11||274||
02-17-2017, 12:22 AM
Last Post: Ghost1958
|ResMed AirCurve 10 ST-A replacement for Respironics System One biPap AVAPS||OldPolioLady||2||134||
02-07-2017, 09:37 PM
Last Post: trish6hundred
|ResMed S9 Humidifer 5i Error Message||atta_gurrrrl||4||2,721||
02-05-2017, 07:56 AM
Last Post: gjw
|[CPAP] Respironics REMStar System One Series 60 Repair||lempface||6||267||
02-04-2017, 07:01 AM
Last Post: cate1898
|SD Card Error (Nothing new)||justMongo||3||158||
02-02-2017, 10:22 AM
Last Post: Crimson Nape
|[Equipment] PR System One Pro 650 - Continuous recycle||spyral||3||149||
01-29-2017, 09:19 PM
Last Post: Sleeprider