Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
Forgot to Change Mask Settings on ResMed S 9
11-23-2013, 05:02 PM
Last night was a bit hellish because I tried out my new Nano and discovered I was a mouth breather big time and I couldn't fix it with a chin strap. So I went back to my ff mask without changing the settings in the machine. I did not do well and my AHI and leakage was higher. How do mask settings affect pressure?
11-23-2013, 05:29 PM
Mask setting might affects unintentional leak (mask plus mouth leak) as each mask type have different intentional leak rate (vents leak) at different pressures
Pressure vs. Flow Rates for ResMed Masks
11-23-2013, 07:11 PM
In other words, as far as treatment goes, no, makes little difference. What matters is how it reports the leak rate. Probably not much of a difference between pillows and mask but sure would be for any other mask type and a full face mask.
Apnea Board Moderator
Breathe deeply and count to zen.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
11-24-2013, 04:04 AM
(11-23-2013, 05:02 PM)Lukie Wrote: Last night was a bit hellish because I tried out my new Nano and discovered I was a mouth breather big time and I couldn't fix it with a chin strap. So I went back to my ff mask without changing the settings in the machine. I did not do well and my AHI and leakage was higher. How do mask settings affect pressure?
Mask setting being wrong would not affect delivered mask pressure.
Mask setting being wrong would cause errors in calculation of Intentional Leak (through mask vent holes) versus Unintentional Leak. This does not affect therapy pressure, only causes error in reporting Intentional Leak versus Unintentional Leak.
Membership in the Advisory Member group should not be understood as in any way implying medical expertise or qualification for advising Sleep Apnea patients concerning their treatment. The Advisory Member group provides advice and suggestions to Apnea Board administrators and staff on matters concerning Apnea Board operation and administrative policies - not on matters concerning treatment for Sleep Apnea. I think it is now too late to change the name of the group but I think Voting Member group would perhaps have been a more descriptive name for the group.
11-25-2013, 10:47 PM
I suspect it may affect things like EPR, bilevel and how quickly the machine regulates the pressure as you inhale and exhale. It might also slightly affect the measurement of AHI and central vs. obstructive.
However, I think the above effects are relatively minor.
11-26-2013, 10:41 AM
Sorry to hijack this thread but does anyone know the difference between nasal and nasal ultra on the mask settings for the S9?
11-26-2013, 01:21 PM
(11-26-2013, 10:41 AM)JohnNJ Wrote: Sorry to hijack this thread but does anyone know the difference between nasal and nasal ultra on the mask settings for the S9?ResMed seem making things simpler (maybe not) with the S9 mask settings as lumping all masks of the same types together, for example "Nasal" for all nasal masks but with two exceptions:
1- Ultra Mirage nasal mask ... mask setting: Nasal Ultra
2- Swft FX Nano nasal mask ... mask settings: Pillows
|Possibly Related Threads...|
|Sudden Unexplainable Change||scottdooley||47||1,603||
8 hours ago
Last Post: Sleeprider
|Change pressure or new mask?||DavdM||2||99||
04-22-2017, 03:42 PM
Last Post: OpalRose
|Advice re: changing CPAP settings||fyrfyter43||46||1,632||
04-22-2017, 12:54 PM
Last Post: fyrfyter43
|Bought New BiPAP - Settings ideas||holden4th||4||165||
04-21-2017, 08:53 AM
Last Post: Sleeprider
|Newbie - need advice on settings and machine||tmc712||21||1,068||
04-20-2017, 05:20 PM
Last Post: quiescence at last
|How often to change P10 pillows||Cranberry Ray||19||1,189||
04-14-2017, 12:57 AM
Last Post: Hydrangea
|Help! Changed CPAP Settings, Now Machine Not Working||cj789||9||380||
04-09-2017, 05:18 PM
Last Post: Ockrocket