Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
Help with Intellipap
08-25-2016, 07:22 AM
i have had my Devilbiss Intellipap autoadjust for several years and love it. last week pulled it off nightstand during the night, landed upside down with water spilling. still turns on but blows at a maximum rate, too much to be ale to wear mask. I went thru clinical settings and lowered max to 10 and then to 8 but no difference in the way its blowing out. also when I exhale into mask (nasal pillow) the preasure doesn't back off, so result is I just have an air hose blowing at a high rate, any ideas on how I can get it back to normal? thanks, robb
08-25-2016, 08:37 AM
Water likely got into the airflow channels of the blower unit and damaged a sensor (probably the pressure sensor.)
So, the system is running open-loop.
There's really not anything to be done as far as self repair or settings.
Supplier #27 and #28 on the supplier list (top of page) do repairs; you might give them a call.
Most often it's more cost effective to just get a new machine with the latest technology.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
08-25-2016, 09:17 AM
jusmongo, thanks so much for the info, I figured that was probably the case, not sure I really need the humidification anyway, may just run without to avoid a repeat
08-25-2016, 09:21 AM
Try setting it out in the sun for a couple of days. If it's moisture on the processor board, you might get lucky. A vacuum chamber would be better but hard to source.
The old method of putting it in a baggie with a quantity of rice might also draw the moisture out. I've never used that system but have heard that it works.
08-25-2016, 05:29 PM
WELCOME to the forum.!
I'm sorry to hear about your machine.
Good luck to you on your machine decision, hang in there for more responses to your post.
|Possibly Related Threads...|
|DeVilbiss IntelliPAP 2 Pulse Dose and Data Questions||Snoopy.pa30||12||302||
6 hours ago
Last Post: analog
|DeVilbiss Intellipap AutoBilevel DV57 Clinical Set Up||JessieLC||5||510||
07-11-2016, 12:06 AM
Last Post: DeepBreathing
|[Equipment] reviews of Resmed S9 AutoSet vs. DeVilbiss IntelliPap AutoAdjust||cat_lover||20||9,021||
05-31-2016, 10:15 AM
Last Post: NCZzzz
|Clinician mode for Intellipap 2||AZCraig||2||742||
01-17-2016, 12:50 PM
Last Post: vsheline
|Intellipap Autoset data cable/adapter?||ghmerrill||2||760||
09-04-2015, 04:49 PM
Last Post: trish6hundred
|Intellipap blowing too much air||MissApnea2000||3||2,084||
06-12-2015, 10:12 PM
Last Post: OMyMyOHellYes
|DeVilbiss Intellipap Auto CPAP Machine with SMARTFLEX (CFLEX) +Sleepyhead||Mark Douglas||0||1,240||
04-21-2015, 09:12 PM
Last Post: Mark Douglas