Thanks for all your help. Seems there is a difference between the S8 and S9 where the S8 raises the pressure after EPR without an inhale.
As I said I shall discuss with supplier and Dr.
in the mean time, (and mostly irrelevant) I finally got my ResMed output to fit into the attachment.
What you cannot see here, is the pressure never varies from 10cmH2O.
This may be that the EPR isn't working (and of course it may biot be set correctly) but I think it is.
The "patient" setting doesn't seem too allow any change by the patient. I'm hoping that it might "learn" from me and get it right. Otherwise I'll go back to my S8 and not be out for sooooo long.
Other than keeping the airway open and collecting lots-o-data what good is a CPAP if it doesn't get me berthing again?
Maybe I'll try a higher pressure (if I can keep the mask on)
Unlike SleepyHead, ReScan does not show IPAP and EPAP separately, cannot see leak graph
If were me, I would turn off EPR and see what happens tomorrow, does not look the set pressure is optimal
Maybe show those results to your doctor
03-10-2014, 12:12 PM
(This post was last modified: 03-10-2014, 12:15 PM by KC2684.)
How in the heck do I delete an attachment. I'm sure the instructions are easy to find but....
OK, please forget this request. It was right there in front of my eyes.. perhaps my mask was in the way.