I download data into both Rescan and Sleepyhead. Just recently I have noticed a disparity in the max leak rate. In Rescan my leak rate is: median 0, 95% 0, max 6 and in Sleepyhead it is median 0, 95% 0, max 15. (Even an old Mirage Quattro does OK)
I just chased it down and it looks like a Rescan bug. The Leak rate graphs in both Sleepyhead and Rescan agree with the Sleepyhead max of 15. I am using an older version of Rescan 126.96.36.19931 and I just converted to the latest and greatest Sleepyhead testing, 64 bit, open GL, within the last couple of days.
Just thought that it was kind of interesting.
03-17-2015, 01:11 PM
(This post was last modified: 03-17-2015, 01:12 PM by justMongo.)
Doesn't ResScan attempt to report unintentional leak rate while Sleepyhead reports total leak (unintentional plus mask vent rate) ???
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
03-17-2015, 03:50 PM
(This post was last modified: 03-17-2015, 03:52 PM by PaytonA.)
I think that you are thinking of PR machines. I am using an S9 VPAP Auto kind'a like what you have. The machine reports the calculated unintentional leak rate and, to my knowledge, that is what both Rescan and Sleepyhead use.