I'm still learning all the lingo and terms so bear with me as I try to explain this.
When I start using my CPAP for the evening, I'm quickly able to adjust/sync my breathing rate with the machine's (or the machine to mine, as it were). At the end of my exhalation cycle, the machine begins to blow air through my mask. However, I have noticed recently that it stops blowing air about 2/3 - 3/4 of the way through my inhalation cycle. In other words, as I'm inhaling, the air stops blowing before I have stopped inhaling.
Is this a function of the pressure setting? My setting is set at 6 by the sleep clinic, and in my first follow up (I've been on CPAP for approx. 4 weeks now), the doctor decided to keep my pressure at 6 for the next two months until my next follow up in March.
This is not really a big issue, it's not like it's keeping me awake (not yet), but I'm just wondering if there's a better way to ensure that the air continues to blow throughout the period that I am inhaling.
If the machine can sense my inhalation/exhalation cycles, then it should be able to sense the entire time I'm inhaling, correct?
01-29-2015, 08:26 AM
(This post was last modified: 01-29-2015, 08:28 AM by quiescence at last.)
I believe this is part of the C-Flex function to create an environment of easing pressure as you begin the exhale. It draws conclusions based on your past flow, most recent breaths, not only the current breath.
As you relax and enter sleep, all things become more steady, and the expected exhale is nearer to exactly when you do exhale. So, ease your mind, during sleep the machine will not be "taking control".
by the way - you can see those times when your inhales and exhales are erratic in the sleepyhead charts, most prevalent by wide swings in tidal volume, signaling periods of wakefulness (or in some cases potentially REM sleep).
Dedicated to QALity sleep.
You'll note I am listed as an Advisory Member. I am honored to be listed as such. See the fine print - Advisory Members as a group provide advice and suggestions to Apnea Board administrators and staff concerning Apnea Board operation and administrative policies. Membership in the Advisory Member group should not be understood as in any way implying medical expertise or qualification for advising Sleep Apnea patients concerning their treatment.
I'm not really that familiar with the PR machines...
(So, take this with a grain of salt...)
My machine has a TImax setting. The Inhalation (IPAP time) is limited by TImax.
So, for me, if it were set too short, I would experience the same effect you describe.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.