Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
04-19-2015, 02:40 AM
I use a ResMed Elite S9 at a pressure setting of 19 and I've noticed that after about an hour the pressure drops significantly causing me to wake up, is this normal? Is this something I can adjust?
04-19-2015, 03:25 AM
G'day Doug, welcome to Apnea Board.
The Elite should maintain a constant pressure all night. If it does drop suddenly, two things come to mind - 1) the machine is experiencing a malfunction, or 2) there is a significant leak. A pressure of 19 is very high - almost at the upper limit of what the machine can produce, so it's not impossible it's overheating. But more likely you're experiencing a leak somewhere along the system.
I see you're not using software - I'd suggest you install Sleepyhead and monitor what's happening. http://www.sleepfiles.com/SH2/ Sleepyhead is user-friendly (to quote an old term) and will indicate the pressure, flow rates, leaks etc. In the absence of software, you can also check the LCD display - press the two silver buttons together for 3 seconds. Then use the push-dial to select and click on the second icon from the top, which will give you a detailed list of your parameters including 90% leaks. This might give you a clue as to what's happening, but the software gives a much more comprehensive report.
Apnea Board Moderator
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
04-19-2015, 10:27 AM
WELCOME! to the forum.!
Hang in there for more suggestions and answers to your question and much success to you with your CPAP therapy.
|Possibly Related Threads...|
|[Treatment] Can a Sleep Number bed change my treatment?||doickle||18||451||
02-21-2017, 11:48 AM
Last Post: Sleeprider
|How often to change P10 pillows||Cranberry Ray||13||659||
02-10-2017, 03:07 PM
Last Post: dogmuck
|how to change pressure setting in XT Fit||Wolfpackfan||6||11,518||
01-25-2017, 08:53 PM
Last Post: johnnyflhx
|Change p10 to n20 mask||Rcgop||9||478||
01-18-2017, 10:20 PM
Last Post: Sleeprider
|[Treatment] Should I change machine||jm54||4||418||
01-18-2017, 07:06 PM
Last Post: kwhenrykerr
|[Pressure] Change Pressure on AirCurve 10 VAuto||info||32||6,208||
12-20-2016, 12:26 AM
Last Post: _cy_
|[Pressure] How to change pressure - Respironics Solo||slgoltz||5||20,904||
11-30-2016, 06:32 PM
Last Post: randx2