Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
[Pressure] Time to change pressure setting?
12-23-2015, 04:02 PM
One month after Respironics System One, a typical night Pressure graph is shown. My Rx is 7.0 but this shows a saw tooth almost every night. (APAP 7-10). Is it time to increase the lower pressure?
The answer to your question depends on your AHI and breakdown of, like obstructions, hypopneas, clear airways, flow limitations, RERA's, and of course how you feel.
You will see that "saw tooth" regardless of pressure. The PR machines do this thing called "hunt and peck" during the night. If you look closely, you will see that it raises the pressure by 1.5cm every so often and drops back to start pressure...hence looking like a sawtooth. It does this to check for events.
What you should post here in addition to the pressure graph is the event graph, flow rate graph, and leak rate graph. Also include the detailed info to the left of graphs.
How to Organize and Post ScreenShots
12-24-2015, 05:03 PM
Your pressure graph looks normal, but a good reason to raise the Min Pressure setting would be if 7 cm H2O is uncomfortably low.
Membership in the Advisory Member group should not be understood as in any way implying medical expertise or qualification for advising Sleep Apnea patients concerning their treatment. The Advisory Member group provides advice and suggestions to Apnea Board administrators and staff on matters concerning Apnea Board operation and administrative policies - not on matters concerning treatment for Sleep Apnea. I think it is now too late to change the name of the group but I think Voting Member group would perhaps have been a more descriptive name for the group.
12-28-2015, 07:29 PM
(12-23-2015, 04:48 PM)OpalRose Wrote: The answer to your question depends on your AHI and breakdown of, like obstructions, hypopneas, clear airways, flow limitations, RERA's, and of course how you feel.
Dear Opal Rose - thanks for your reply. Here are the additional graphs. AHI is very low compared to others, but still feel tired.
12-28-2015, 08:15 PM
Your graphs look ok, a high leak issue around 3:00. Was looking for your medium and 90% pressure numbers. If you check your 90% pressure, (this means you were at that pressure for 90% of the night and under, then your starting pressure should start 1 or 2cm below the 90% number.
You probably are starting at the right pressure, but could be raised to 7.5 or 8. If you do this, then start your ramp higher, around 6. You can also use a flex setting, try at 2, this will help with exhale relief.
You just may be sensitive to pressure changes, and may be waking you up, but you haven't been on Cpap for very long, so give it some time.
How to Organize and Post ScreenShots
|Possibly Related Threads...|
|Change p10 to n20 mask||Rcgop||9||308||
01-18-2017, 10:20 PM
Last Post: Sleeprider
|[Treatment] Should I change machine||jm54||4||201||
01-18-2017, 07:06 PM
Last Post: kwhenrykerr
|newbie battery question- time sensitive||Dogwood1447||14||377||
01-18-2017, 07:06 AM
Last Post: pc1955
|How to ulock Ramp setting on dreamstation||wdriver||3||189||
01-15-2017, 09:15 PM
Last Post: wdriver
|What's the maximum pressure setting that doesn't disturb your sleep?||Ihaveapnea||12||520||
01-14-2017, 10:02 AM
Last Post: Newbee2016
|What is the expected time for continuous sleep?||Benji||10||500||
01-13-2017, 12:50 PM
Last Post: Mosquitobait
|Time Required to Become Accustomed to ASV Machine||Southerngent1||8||230||
01-10-2017, 05:11 PM
Last Post: robertbuckley