Hello Guest,Welcome to Apnea Board !
Quick Setup question for PRS1
07-31-2016, 07:43 AM
I can only speak for myself, but my System One will hike the pressure up to whatever max I set if it determines Flow Limitations and Snores, which in my early treatment I experienced many FL, Snores, and RERA's. And that hike in pressure would ultimately wake me up, so I keep my max setting to 2cm above my 90% number. On occasion, I encounter FL and Snores that will send the pressure right up to my maximum setting of 13, with no other apnea events in site, and I don't want it going any higher if not needed. Hope that makes sense.
How to Organize and Post ScreenShots
07-31-2016, 08:02 AM
Makes total sense.
07-31-2016, 09:51 AM
Hi, cate. This thread has some discussion of reasons for capping max pressure. See post(s) by vsheline in particular.
(07-31-2016, 06:10 AM)cate1898 Wrote: Second question, why do we cap off the upper range in auto mode? I understand setting the lower end of the range and why it is better, but if your pressure is only going to go as high as needed then why not just leave the upper at 20? Maybe I'm just confused?
07-31-2016, 12:27 PM
Thanks green wings! It's good I guess to avoid the possibility of increased Centrals and increased Intraoccular pressure. Good catch!
APNEABOARD - A great place to be if you're a hosehead!!
EVERY ACCOMPLISHMENT BEGINS WITH THE DECISION TO TRY!
These machines have algorithms in them that are tuned for the "average" user, you might say. But none of us are "average". While the algorithms are pretty good for a lot of people they do not work for everyone. If I leave my machine open at the top it goes pretty close to 20 and seemingly tries to blow the mask off my face. BUT it doesn't make my AHI any lower. In fact at a steady pressure of 11 with EPR set at three my AHI is pretty well always under one and zero maybe half the time. I'm not the patient this particular machine was designed for, although it works just fine as a CPAP machine and keeps my AHI low. But a low AHI doesn't help me much on its own because my 02 levels on the machine alone go very low even though no or very few apneas are recorded, obstructive *or* central. So they added oxygen in during sleep and that made the difference. I still want the positive pressure to prevent the obstructive events and to get good data on how I am doing.
Of course I am probably an outlier with very unusual requirements, but still since most of us aren't average it makes sense to me that many of us will be better off without the machine set to run "wide open".
I am neither a Doctor, nor any other kind of medical professional.
Actually you know, it is what it isn't.
07-31-2016, 09:10 PM
X1 will work for any ffm or nasal mask.
As for pillows id set to 0. There are several x settings for different pillow masks in the repironics line. Id doubt you would hit the right pillow setting guessing.
The x setting has nothing to do with leak rate, everything to do with machine overcoming mask resistance to flow and pressure.
Use O on non respironics pillows
|Possibly Related Threads...|
|No setup menu option RemStar Pro 460P||buschjd||9||187||
01-11-2017, 05:58 PM
Last Post: OpalRose
|HELP! machine died and need a quick solution!!||Jamer in trouble||12||767||
12-19-2016, 07:04 AM
Last Post: E.W.
|First machine!! Just arrived, setup help please.||SJGino||4||357||
11-16-2016, 08:49 PM
Last Post: trish6hundred
|My battery backup setup||GeoffD||278||23,894||
10-27-2016, 05:28 PM
Last Post: Sleeprider
08-16-2016, 07:46 AM
Last Post: OMyMyOHellYes
|New, quick questions||KarKat||6||422||
08-04-2016, 04:31 PM
Last Post: KarKat
|DreamStation Auto - setup question||KSMatthew||3||455||
07-11-2016, 11:18 PM
Last Post: chill