Hi all I hope someone can give me some help with this. Just downloaded the RESScan software and loaded my card up. I was wondering if some once can explain the leak part of this. For example on October 25th when I hover over the green bar it reads:
I'm thinking I have a leak problem which is maybe why I constantly wake up with dry mouth and I don't feel rested.
can somebody help?
140 L/m ... maximum leak ... highest leak detected
0 L/m ..... 95% percentile leak .... leak been at or below this number for 95% of the time
0 L/m ..... median leak ...50% percentile leak ... leak been at or below this number for 50% of the time
You have zero leak for 95% of the time, the higher leak rate maybe due adjusting mask fit or/and keeping the machine on while taking a pee during the night . S9 consider good mask fit when the 70% percentile leak is less than 24 L/m
You can try increase humidity a little, see if helps with the dry mouth. Have you got the climate line (heated hose), the climate line on auto mode allows you to adjust temperature to preference with rainout protection