From another S9 user;
As my hubby (a WiFi tech guy) always tells me "When something goes funny - REBOOT!"
So, unplug the S9 for about 5 minutes, at least. Plug it back in, test it during the day, as retired_guy suggested, and see how it does.
Also, as retired_guy suggests, I 2nd the idea of downloading the SleepyHead software.
Evpraxia in the Pacific Northwest USA
Diagnosed: 44 AHI when supine, O2 down to 82%
Treated since 20 Sept 2014:: 0.7 AHI, Settings 7-15, EPR on Full Time at Level 3
Better living through CPAP/APAP machines!
[quote='retired_guy' pid='103614' dateline='1425754773']
Thank you retired guy for responding to the problem I posted. I was able to follow your instructions, unpluging the machine, making the changes in settings, using a different hose and mask to take that out of the equation, and got in a very comfortable chair for a 15 minute rest.
There may have been a little difference from before, but I still felt strong pressure when breathing in and it took effort to exhale.
Just to check, I took the mask and hose to my other unit and powered it up. The difference was dramatic. I felt assistance in inhaling, but not strong pressure, and exhaling was very easy.
Not sure if something else to try, but from the experience, I am now sure there is a problem.
03-07-2015, 10:45 PM
(This post was last modified: 03-07-2015, 10:47 PM by vsheline.)
(03-07-2015, 11:37 AM)Jks789 Wrote: My machine is a couple years old.
The clinician set-up manual (see link at top of all forum pages) will say how to Reset Factory Defaults. If that does not fix the problem then the blower needs repair or reacement.
If you received your machine less than 24 months ago bring it back to the provider immediately to be replaced under warrantee.
If you received it between 24 month to 27 months ago, you can call ResMed Customer Service and give them the REF# and Serial# on the back of the blower unit, and they will tell you whether is is still under warrantee. ResMed warrantees machines for 2 years plus 3 months from date of manufacture, which they will look up based on the serial number.
There may be a chance that you can have the unit replaced for free under the 27 month warrantee which started on date of manufacture.
If you received it longer ago than 27 months, it may be better to buy a gently used S9 AutoSet from Supplier 2 or from Craig's List. A link to the Supplier List is at top of every forum page. If dealing with Craig's List, be very careful, and also you might not want to buy a machine from a smoker. Usually best to discard or clean very well the water tub and hose and mask.
Membership in the Advisory Member group should not be understood as in any way implying medical expertise or qualification for advising Sleep Apnea patients concerning their treatment. The Advisory Member group provides advice and suggestions to Apnea Board administrators and staff on matters concerning Apnea Board operation and administrative policies - not on matters concerning treatment for Sleep Apnea. I think it is now too late to change the name of the group but I think Voting Member group would perhaps have been a more descriptive name for the group.