Hello Guest,Welcome to Apnea Board !
ResMed s8 elite II data
01-05-2014, 07:30 PM
When I look at data on my machine for the week am I looking at data for last 7 days or for the calendar week ? And is this an average or median ?
Sleep Well !!!
01-05-2014, 08:12 PM
01-07-2014, 11:16 AM
(01-05-2014, 07:30 PM)mjpr Wrote: When I look at data on my machine for the week am I looking at data for last 7 days or for the calendar week ? And is this an average or median ?
...And the last 7 days. It has a clock, but no calendar, so it has no idea what a "calendar week" is.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
01-07-2014, 05:18 PM
Correction, from S8 manual
Efficacy Data: Displays pressure, leak, AHI (apnea + hypopnea index), AI (apnea index), and HI (hypopnea index) data for a range of time intervals.
Usage Data: Displays usage data (including Run Hours) for a range of time intervals
Data Management Using the Menus
The S8 AutoSet II system may be used to monitor patient usage as well as treatment pressure, mask leak, and incidence of apneas and hypopneas. To assess the patient’s progress, data for the last session may be compared to median values for the last week, the last month, the last six months, and the last year, as displayed in the Results menu. The device stores usage and summary data for up to 365 sessions.
The USED screen reports total hours the device has been used and the number of days the device was used compared to the total days since the hour meter was last reset.
Other statistics quoted include:
• pressure (95th centile value for mask-on time only)
• leak (95th centile value for mask-on time only)
• AHI: apnea + hypopnea index (calculated per session)
• AI: apnea index (calculated per session)
• HI: hypopnea index (calculated per session).
|Possibly Related Threads...|
|ResMed S9 SD Card "Invalid SD Card, erase card data?"||MAPnea||31||29,880||
03-14-2017, 09:34 AM
Last Post: Sleeprider
|Airsense 10 Elite humming all night - going crazy!||Cro0ked||16||861||
03-01-2017, 07:55 PM
Last Post: nanbar
|[News] IS RESMED SELLING DATA TO DR. OZ?||srlevine1||11||587||
01-11-2017, 03:59 PM
Last Post: tedgreen
|[Equipment] Auto start on resmed elite s9||justka||5||354||
01-08-2017, 07:42 PM
Last Post: zonk
|Why does Sleepyhead data not match my machine's data?||Ihaveapnea||8||617||
01-06-2017, 06:19 PM
Last Post: Ihaveapnea
|Need to get Data from Resmed Vpap Auto (not the S9)||wilkr||3||214||
01-05-2017, 12:10 AM
Last Post: justMongo
|[CPAP] No data on a Resmed S8||Kozyair||15||1,414||
01-04-2017, 05:06 AM
Last Post: OmgLetMeSleep