Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
Resmed Leak displayed on S9 Elite
09-12-2013, 12:51 PM
I have been searching this forum and Resmed sites to find out what is displayed as Leak in Lpm on the S9 Elite. Is it intentional leak only, through the mask vent holes, unintentional (mask leak) or total leak, intentional plus unintentional. I have been reading leaks of 38 to 41 Lpm with Resmed swift fx and Quattro Mirage masks since I got this new machine. Sure would appreciate a definitive response.
09-12-2013, 01:32 PM
I believe it is total leak.
Have you gotten the clinician manual yet?
I just glanced through it and it doesn't say what it is. Given that they keep it simple, I would assume total leak so no one has to do math.
Apnea Board Moderator
Breathe deeply and count to zen.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
09-12-2013, 01:37 PM
WELCOME! to the forum.!
Hang in there for more responses to your post and best of luck.
09-12-2013, 04:40 PM
(09-12-2013, 12:51 PM)steveh Wrote: I have been searching this forum and Resmed sites to find out what is displayed as Leak in Lpm on the S9 Elite. Is it intentional leak only, through the mask vent holes, unintentional (mask leak) or total leak, intentional plus unintentional. I have been reading leaks of 38 to 41 Lpm with Resmed swift fx and Quattro Mirage masks since I got this new machine. Sure would appreciate a definitive response.Unlike other machines report total leak ... ResMed machines does the calculation by subtract intentional leak and reporting unintentional leak
(mask plus mouth leaks), for this reason you've to select the type of mask you,e using as different masks have different vents leak rate (intentional leak). ResMed consider leak at or below 24 L/min acceptable leak, higher leak rate may affect the accuracy of the reported data so good idea to maintain leak below the red line in ResScan leak graph. Chinstrap can helps to remind us keep mouth closed and minimize mouth leaks
|Possibly Related Threads...|
|Airsense 10 Elite humming all night - going crazy!||Cro0ked||13||573||
02-11-2017, 06:05 PM
Last Post: Cro0ked
|AirFit P10 leak or no leak?||ppca||4||287||
01-18-2017, 01:50 AM
Last Post: ppca
|[Pressure] Another New Player with Some Leak Issues||Platypus||8||558||
01-16-2017, 10:45 AM
Last Post: Cranberry Ray
|[Equipment] Auto start on resmed elite s9||justka||5||273||
01-08-2017, 07:42 PM
Last Post: zonk
|Acceptable Leak Rate for WISP Nasal Mask||bensonlee||6||339||
12-31-2016, 03:33 AM
Last Post: trish6hundred
|One good night, now leak issues?||sleepyguy99||24||1,496||
12-28-2016, 02:08 PM
Last Post: sleepyguy99
|[Equipment] Water leak risk from ResMed AirSense 10 Humidifier tank||ac110||8||592||
12-18-2016, 06:20 PM
Last Post: PoolQ