07-24-2015, 10:47 AM
(This post was last modified: 07-24-2015, 10:48 AM by PaytonA.)
What is it that you are calling the "Breathing element"? Are your tanks 2 piece or three piece?
I use a little more than half the tank each night. Almost 2/3rds. My usage is about 8 hours. My Airsense 10 humidity is set to 4. We run AC all the time and the rH is on the dry end of normal.
Using FlashAir W-03 SD card in machine. Access through wifi with FlashPAP or Sleep Master utilities.
I wanted to learn Binary so I enrolled in Binary 101. I seemed to have missed the first four courses.
I tested the humidity on my Airsense 10 for a few days before turning it off again. Setting was 4. This was in April, so no air conditioning was on yet. It used 2/3 of the tank each night. I slept for about 8 hours.
I work for a DME and from what I have seen about 1-2% of the A10 water chambers have a leak thru the sonic welds on the bottom of the tanks. They only start leaking once the pressure is on and since it is from a weld you won't see any cracks. As for the amount of water used each night that can vary widely with the RH and temperature of the air from the room. Another problem the tanks have is there were several lots that had a manufacturing defect and they will make a whistling sound when the machine is in use.
Thanks for the intel.