(02-18-2014, 05:52 PM)3porpoise Wrote: Hi,ResMed define acceptable unintentional leak at 24 L/m and prolong high leak above that number the machine cannot compensate or give accurate data reading. 95the percentile means leak was at or below 30.6 L/m for 95% of the time (not 30.6 L/m all night night long), of course the lower the better but because median leak is low at 2.4 L/m, leak in this case is not a problem (might be just adjusting mask fit, scratching/blow the nose, or even leaving the machine on while taking a pee which is better to turn machine off) as long as not getting close to the eyes or disturb my sleep
If I'm not mistaken , believe 95th percentile leak of 30.6 is too high is it not? Other more experienced members can comment on this but should it not be 24 or less ?
Can someone verify this please?
Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
|Possibly Related Threads...|
|Maximum CA number||CPAPreturnee||13||253||
05-21-2017, 12:08 PM
Last Post: CPAPreturnee
|Theoretical Setting for a Bi-Level Machine Question||Crimson Nape||1||88||
05-18-2017, 08:10 AM
Last Post: Sleeprider
|Mistake setting pressure||sleepytimegal||12||401||
05-16-2017, 10:16 AM
Last Post: sleepytimegal
|Thoughts on my Auto Data?||Hydrangea||18||490||
05-07-2017, 10:50 PM
Last Post: Hydrangea
|Periodic breathing, and higher pressures||Hockenberry||8||328||
05-02-2017, 05:30 AM
Last Post: ajack
|Airsense 10 hybrid mask setting||TexasTiger||7||294||
04-30-2017, 02:37 PM
Last Post: TexasTiger
|resmed n20 mask setting||Saxmann||5||187||
04-26-2017, 12:40 PM
Last Post: Saxmann