WELCOME! to the forum.!
It's good to hear that you are restarting CPAP therapy and that you got a different machine.
Hang in there for more responses to your post and much success to you as you restart your CPAP journey.
What is your model of S9. Is it an elite, autoset, etc.?
It would be a good idea to put S9 Autoset in your profile to avoid a lot of people asking the same question.
07-07-2016, 06:22 AM
(This post was last modified: 07-07-2016, 06:23 AM by OpalRose.)
It looks like you had a couple mask leaks. You may want to try mask liners to help with that.
Also, be sure your not over tightening the straps.
Your data doesn't look bad, but the minimum start pressure of 4 is too low. Your profile says 6-20, so you maybe changed it which is good. Try that for awhile, but I think you may need a slightly higher pressure, possibly 7 or 8.
I doubt you need to keep the max pressure at 20. Bring it down to 15.
Of course, this is one night, so watch your events. Would like to see the hypopneas and Obstructives come down, and a little higher minimum pressure may take care of that.