Well I finished my first week on CPAP with a bang. I looked at AH this morning and found a little area where I looked like I died. lol. Really tho, looks to me like I had a really large leak. I couldn't use a FFM so Until I can get this P10 changed out (really hurts my nostrils) and try a Dreamwear, I use medical paper tape and tape my mouth shut. Looks like I blew it off, found it when I got up.
I also had a lot of events at first cuz I had the mask on while I was getting set up ie taping, moving blankets, getting my SPo2 on and so forth. Then I slept for 3.5 hours, woke and fell back asleep for 2.5. Got up, took off the mask and slept another hour and woke after having a really bad dream!
So that was it. Is this how SH interprets a big leak? The P10 really doesnt work that well for me cuz I cant get it comfy on my nostrils. I wake up leaking at times and if I tighten it up more, it hurts.
A couple tips on the P10 you may already know:
Straps close together at the crown of your head = tighter
Straps spread apart = looser
Large pillows are better than smaller for irritation. You want the seal at the nares.
The P10 works best when not overly tight. You should be able to lift it away from your nose and comfortably set it in place and get a comfortable seal. If it is mashing your nose, you're doing it wrong.
You seem to have had a displacement leak at 00:45, but the leak at 01:00 looks like a mouth leak. Everything else is acceptable except 03:15, so it's not that bad for a new user in the first week. You're doing better, and irritation should start to wane.