Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
clearing s9 memory
02-05-2013, 02:34 PM
I bought a used autoset and coincidentally the first user started with Feb 2 of 2012, so our stats kind of overlap - is there any way to clear that first users data out?
02-05-2013, 03:03 PM
you need to go to the clinical menu (Configuration) to erase the data
allows you to erase all data stored
machine hours cannot be erased
the manual is available via email (scroll down to section three)
S9 setup information and video
02-05-2013, 03:08 PM
Yes, in the clinician menu under configuration. It will erase all but the total hours used (machine hours). If that is important to you, then just write the hours on a sticker and put on bottom of machine.
Do you have the clinician manual? You should get it if you don't.
Hold the silver button that has the two checkmarks on it while also holding down the knob. Let go when the screen changes to the clinician menu.
Turn knob to select the "Configurations" menu.
Tap knob (screen turns orange I think)
Turn knob to select "Erase data".
I would assume you then tap the knob again. Not sure what the screens say after that.
Apnea Board Moderator
Breathe deeply and count to zen.
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
02-05-2013, 03:19 PM
Got it - thanks! Now to see MY data tomorrow.
02-05-2013, 04:25 PM
02-06-2013, 01:53 PM
02-11-2013, 12:20 AM
So, how do we do a full reset? I am thining that my S9 Autoset is not blowing what it should be blowing; the pressure appears to be lower than indicated. I compare it to my spare set to the same pressure and the spare appears to be blowing a LOT more pressure than the S9 Autoset. I'd like to do a full hard reset. Next I'd like to source a manometer and actually measure the pressures. Anyone have a manometer they'd like to sell?
Educate, Advocate, Contemplate.
Herein lies personal opinion, no professional advice, which ALL are well advised to seek.
02-11-2013, 01:21 AM
Supplier #2 have one for $35 include shipping
|Possibly Related Threads...|
01-08-2017, 07:25 AM
Last Post: kwhenrykerr
11-25-2016, 02:15 PM
Last Post: kwhenrykerr
|[CPAP] Do sleep MD's use memory card data?||beet||15||1,006||
09-23-2016, 07:25 PM
Last Post: richb
|Clearing rain out from tubing||shokk||13||1,042||
07-07-2016, 04:21 AM
Last Post: shokk
|[Equipment] what kind of memory card is used for REMSTAR System One DS560?||asdftalk||10||763||
07-03-2016, 09:18 PM
Last Post: egzbuen
|Machine not clearing info at noon||me50||4||954||
09-02-2015, 09:17 PM
Last Post: me50
|ResMed memory card||pgstegall||20||3,391||
03-08-2015, 08:16 PM
Last Post: archangle