Hello Guest,Welcome to Apnea Board !
As a guest, you are limited to certain areas of the board and there are some features you can't use.
To post a message, you must create a free account using a valid email address. Login or Create an Account
smart start for resmed s9
08-18-2012, 03:31 PM
looking for info on what smart start for resmed s9. what is it's function??
08-18-2012, 03:55 PM
Hi jimmy2708, First of all, WELCOME! to the forum.! If I'm not mistaken, the "Smart Start" function can either be turned off or on. It's function is that you can either have your machine start when you breath into it or if it's turned off, you will have to start your machine. Some people like that and others don't, it's each individual's preference. Hang in there and someone will be along shortly with more suggestions best of luck to you.
08-18-2012, 04:30 PM
Yep, Trish is correct.
From the ResMed S9 manual:
Quote:Note: If your clinician has enabled SmartStart your device will start automatically when you breathe into the mask and stop automatically when you remove the mask.
You may have to enable SmartStart from within the Clinician Menu. You can get directions on how to do that from the Clinician Manual... request a free manual via email by following the directions on this page:
Apnea Board Administrator
INFORMATION ON APNEA BOARD FORUMS OR ON APNEABOARD.COM SHOULD NOT BE CONSIDERED AS MEDICAL ADVICE. ALWAYS SEEK THE ADVICE OF A PHYSICIAN BEFORE SEEKING TREATMENT FOR MEDICAL CONDITIONS, INCLUDING SLEEP APNEA. INFORMATION POSTED ON THE APNEA BOARD WEB SITE AND FORUMS ARE PERSONAL OPINION ONLY AND NOT NECESSARILY A STATEMENT OF FACT.
08-18-2012, 05:20 PM
ok and thanks!
03-18-2013, 04:46 PM
(08-18-2012, 04:30 PM)SuperSleeper Wrote: Yep, Trish is correct.
I'm interested in this as well. Unfortunately when I was using a borrowed S8, I didn't seem to be given the option of activating SmartStart, even in the clinician menu. Does it matter which mask I'm using? When all this waiting game is over, and I have an S9 AutoSet in my grubby little hands, that is one thing I'd like to make use of. But I have no idea if it doesn't care what mask I'm using or what (it's a ResMed Mirage Micro.)
03-18-2013, 07:40 PM
Smartstart may not work if you have a FFM. The anti asphyxia valve may not let enough air go down the hose to trigger the machine to start. It does sometimes work, but it's less reliable.
If you set the mask setting on the machine to FFM, some versions of the S9 firmware will lock out smartstart. I think more recent machines do not do this.
|Possibly Related Threads...|
|Need help changing start pressure airsense a10||Gary1of2||1||94||
05-18-2017, 08:40 PM
Last Post: bonjour
|[How to] Start a New Thread with the Right Subject Line!||PaulaO2||21||2,428||
05-18-2017, 07:45 PM
Last Post: Galactus
|[Equipment] What do I need to do to start using it again???||needmoresleep||5||224||
04-25-2017, 07:07 PM
Last Post: Sleeprider
|Pressure too low to start||inconsistentevents||16||473||
04-09-2017, 05:26 AM
Last Post: trish6hundred
|Why does my pressure start increasing when I'm awake?||Ihaveapnea||11||682||
01-17-2017, 11:17 PM
Last Post: Beej
|[Equipment] Auto start on resmed elite s9||justka||5||484||
01-08-2017, 07:42 PM
Last Post: zonk
|Now THIS is the way to start the new year! AHI of Zero, Zip, Nada last night!||geekbearinggifts||4||352||
01-03-2017, 06:23 AM
Last Post: pupcamper