(03-30-2016, 01:44 AM)robysue Wrote: past the maximum pressure setting?
Please note: I understand that rainout might be mistaken for snoring or a flow limitation and lead to a pressure increase. But we're not talking about a run of the mill pressure increase here: We're talking about the machine increasing the pressure past the maximum pressure setting, and that's very unusual.
In other words, since the max pressure setting is 8 cm, why would the machine decide that it needs to deliver 10cm?
To debby2242: Any chance you show us the detailed data for 3/28/2016? By the detailed data, I mean the the graphs for the flow rate, pressure, and leak rate? You can do this in either ResScan (if you downloaded "all" the data) or SH. I believe that SH 0.9-8 works with the AirSense 10 AutoSets.
I had the same problem and thought it was my machine. I went through three machines - now on my fourth- with this problem. I called Resmed and they said yes the machine CAN do this. I traced it to rainout control and that fixes the problem. I'm pretty sure this is the OPs problem. This is covered in great detail in my original postings on the issue when I had the same problem back last year. I posted some data at the time as well to show it.